The Multi Channel oscilloscope software has a modular structure, with Sources, I/Os and Sinks.
Sources are modules that only generate data. The channels of instruments that are connected to the computer can be used as source, they generate the data that is measured by the hardware. The other source that is available is the software generator, which can create various signals from software that can be used in combination with measured signals.
Sinks are modules that only accept data. Typical sinks are e.g. graphs and meters, they accept data and display it. Other sinks are the disk writer, to store data to disk and the sound sink, to make data audible through the sound system of the computer.
I/Os are modules that accept data (Input), process this data in a specific way and then generate the processed data (Output). Data that is generated by an I/O can be used as input for a Sink or for another I/O. Multiple different I/Os can be chained to each other to perform a complex operation on the data. The Multi Channel oscilloscope software has many different I/Os to process data in a specific way. There are I/Os in several categories:
These I/Os perform a basic mathematical operation on one or more signals:
|Add/Subtract||Add or subtract up to 32 signals|
|Multiply/Divide||Multiply or divide up to 32 signals|
|SQRT||Calculate the square root of a signal|
|ABS||Determine the absolute value of a signal|
|Differentiate||Differentiate a signal|
|Integrate||Integrate a signal|
|Log||Calculate the logarithm of a signal|
These I/Os perform a more complex mathematical operation on a signal or consecutive signals:
|Gain/Offset||Apply gain and/or offset to a signal|
|Average||Outputs the average of n measurements|
|Low-pass filter||First order low-pass filter|
|Min/Max detector||Peak detection with optional fall-off|
|Limiter||Clip the input signal between a minimum and maximum limit|
|Resampler||Changes record length and sampling frequency of an input|
|Data collector||Appends small chunks of data to one big record|
These I/Os analyze the input data and extract specific information from it.
All measurement setups in the Multi Channel oscilloscope software are built from one or more sources, I/Os and sinks. These are simply connected to each other by dragging and dropping. Multiple I/Os can be used to form complex mathematical operations.
In the image above, three I/Os are used to determine the area that is enclosed by two signals in XY mode.