Probe compensation cable TP-DB9-BNC-30
The Probe compensation cable TP-DB9-BNC-30 is a cable that provides easy access to the probe compensation connection at the extension connector of supported instruments.
Order code: TP-DB9-BNC-30
Price: € 14.00
When you want to use attenuating oscilloscope probes like the Oscilloscope Probe 1:1-1:10 - HP-3250I, the probes require a frequency compensation to match them with the input channels. For that purpose, the scopes have a dedicated output on their extension connector, that produces a 1 kHz, -1 V to 1 V square wave signal. The Probe compensation cable TP-DB9-BNC-30 gives an easy access to that signal.
To connect the probe to the probe calibration signal at the extension connector of the instrument, remove the sprung hook from the probe and connect the BNC adapter that is delivered with the probe. Connect the Probe compensation cable TP-DB9-BNC-30 to the extension connector of the scope and the probe with BNC adapter to the BNC connector of the Probe compensation cable TP-DB9-BNC-30 . Make sure to set the probe to its x10 setting and follow the probe compensation procedure.
When your have a Handyscope HS6 DIFF or WiFiScope WS6 DIFF, it requires option SafeGround to be installed and enabled to use attenuating oscilloscope probes.
The table below shows detailed specifications of the Probe compensation cable TP-DB9-BNC-30 .
Probe compensation cable TP-DB9-BNC-30 specifications
|Cable length||30 cm (19.7 inch)|
|Instrument side||9 pin male D-sub|
|Probe side||female BNC|
|Signal||Pin 3||Center pin|
|Ground||Pin 6||Outer shell|
|Compatible instruments||Handyscope HS6 DIFF|
WiFiScope WS6 DIFF